- TECHNICAL SPECIFICATIONS
ASTM D571 Standard Test Methods for Coated Fabrics
ASTM D571 covers, but are not limited to, rubber-coated fabrics, that is, tarpaulins, rainwear, and similar products.
Significance and Use
Coated fabrics and rubber products made from coated fabrics are important items of commerce, and adequate test methods must be available to ensure
the quality of these products.
These test methods provide for this testing capability
ASTM D751 / Testing Machine
The machine shall consist of three main parts:
Clamps for holding specimen, and
Load and elongation recording mechanism(s).
ASTM D571 / Straining Mechanism
A machine wherein the specimen is held between the two clamps and strained by a uniform movement of the pulling clamp shall be used.
Unless otherwise specified in the material specification, the machine shall be adjusted so that the pulling clamp shall have a uniform speed of
5 6 0.2 mm/s (12 6 0.5 in./min).
ASTM D571 / Load and Elongation Recording Mechanism(s)
Calibrated dial, scale, or chart to indicate applied load and elongation.
Unless otherwise specified for load determination, the machine shall be adjusted or set so that the maximum load required to break the specimen will
remain indicated on the calibrated dial, scale, chart, autographic recording, or computer interfaced with the testing machine after the test specimen has ruptured.
ASTM D571 / Capacity
The machine shall be of such capacity that the maximum load required to break the specimen shall be not greater than 85 % or less than 15 % of the rated capacity.
Machine Effıciency—The error of the machine shall not exceed 2 % up to and including a 222-N (50-lbf) force and 1 % over 222 N and 1 % at any reading within its loading range.
ASTM D751 / Clamps for Fabrics
The clamps for holding a specimen of coated fabric shall have gripping surfaces sufficiently flat and parallel as to prevent the test specimen from slipping
or moving between the gripping surfaces when held under the pressure normal to operation.
The dimension of all gripping surfaces parallel to the direction of application of the load shall be 25 mm (1 in.); the dimension perpendicular to this direction shall be
1 in. for the face jaw and 50 mm (2 in.) or more for the other.
All edges that might cause a cutting action shall be rounded to a radius of not over 0.4 mm (0.0156 in.).
The pressure between the gripping surfaces, sufficient to clamp the specimen firmly before the testing load is applied and to prevent slippage during the progress
of the test, shall be secured by any suitably constructed mechanical device operating on the member of the clamp.
The distance between the clamps at the start of the test shall be 75 mm (3 in.)
*** Before conducting ASTM D751 , it is important to read the entire specification. Standards can be obtained from appropriate standard authorities.
***PARSROS offers several types of grips and fixtures which will enable you to perform a variety of tests
that are accurate and repeatable.
Please Contact with our engineers so that we can find and offer Best Universal Tensile Test Machines , Grips , Jaws and Other Accessories for your operations